Patients trust that the water you're using in their oral cavity is of the highest quality. To ensure this, dental unit waterlines must be carefully and consistently maintained. The A-dec line of ICX products helps simplify your waterline cleaning routine.
ICX waterline treatment tablets and new ICX Renew waterline shock treatment work together to eliminate buildups of odor-causing, foul-tasting microbial contamination and to prevent future accumulation.
Ready to use and formulated without harsh chemicals, ICX products are an easy, effective way to maintain your dental unit waterlines and equipment—protecting your investment.
Product availability differs by market - please check with your local distributor.
The effervescing ICX formula does the work for you, preventing the accumulation of odor and foul-tasting bacteria. Simply add one tablet to an empty self-contained water bottle before each filling. ICX continues working in the system with each treatment, protecting the waterlines from further contamination and reducing the need for frequent shock treatments.
Fast-acting ICX Renew liquid shock treatment removes odor and foul-tasting bacteria in dental unit waterlines. Just mix the two solutions together into an empty water bottle, run through the lines, and soak overnight. After completing the shock treatment, flush the lines with water and resume your regular use.
Along with a daily waterline maintenance protocol, it's important to monitor water quality and periodically shock the lines to clear deposits and contamination. These three steps help maintain your dental unit waterlines.
Download the ICX & ICX Renew Dental Waterline Treatment Brochure for additional details.
How long does ICX last in the water bottle or system?
Intended for use with potable water, ICX remains active in the system for up to two weeks.
Can ICX be used in patient's mouths?
ICX has been formulated to be used in water used for dental treatment and thus may be used in a patient's mouth.
Can ICX be used in oral surgery procedures?
The CDC recommends the use of sterile irrigating solutions and devices designed for delivering sterile irrigating fluids for oral surgery procedures. ICX does not create sterile water and thus is not intended for use during oral surgery.
Each ICX Renew application contains two bottles. Do I use both bottles to treat my dental unit?
Yes, combine the entire contents from Solution 1 and Solution 2 in the dental unit water bottle immediately before treating your unit. Do not premix the two bottles to store for later use.
How frequently should I use ICX Renew?
Before you use new equipment for the first time, apply one treatment. For existing equipment, when water quality test results exceed your water quality action level apply three treatments on successive days. You may use the dental unit between treatments. For proactive preventive maintenance, apply one treatment per unit on a routine basis, such as quarterly.
What is the recommended dental unit water quality action level?
Guidelines or requirements for dental unit water quality vary by region and are usually specified in colony forming units per millilitre (CFU/ml). A-dec recommends an action level based on the quality guidelines in your region. The U.S. Centers for Disease Control and Prevention (CDC) recommended guideline is ≤ 500 CFU/ml heterotrophic water bacteria. If guidelines or requirements are not available in your region, A-dec recommends 500 CFU/ml as an action level.
How frequently should I test my dental unit water quality?
The frequency of water quality testing depends on your water quality goal. A-dec recommends initially testing once per month. If monthly results pass your clinic's water quality goals for three successive months, reduce the testing frequency to once every three months. Water quality should be monitored by performing a test that provides a quantitative measurement of heterotrophic bacteria.
How do I test my dental unit water quality?
A-dec recommends using either an in-office water quality test kit or a water quality testing service that provides a quantitative measurement of heterotrophic bacteria.
Which dental unit waterlines should be treated with ICX Renew?
Perform the shock treatment in all air/water syringe and handpiece waterlines that use water within the delivery unit and assistant instrumentation. This includes air/water syringe, ultrasonic scalers, and handpiece waterlines on units that are not currently being used.
How long should ICX Renew be left in dental unit waterlines?
ICX Renew should remain in your dental unit for a minimum of 12 but no longer than 96 hours. See the A-dec Waterline Maintenance Guide and the ICX Renew Instructions for Use for more information on adjusting the application time.
Is it necessary to remove the handpieces, motors, and couplers before applying ICX Renew?
Yes. Remove air/water syringe tips, handpieces, detachable motors and couplers, and ultrasonic instruments from their tubing before adding the solution into the empty water bottle. Clean, disinfect, and sterilize these instruments according to the manufacturer's instructions.
Can I use the vacuum system (HVE/SE) to flush ICX Renew from the waterlines?
No, you must flush ICX Renew into a sink or cuspidor attached to municipal drain.